Five Advantages Of Products
- Without preheating, it can be stably measured in 3 seconds after starting up. Small and portable, switch with reading lock function, under-voltage display and charging indicator function.
- High-gloss standard plate using quartz single crystal material, annual change rate is less than 0.3% , with international advanced level of stability.
- Special Light Bulb Life Up to 10,000 hours; zero point automatic correction, use do not have to zero.
- Using nickel-hydrogen rechargeable battery, its capacity is large (100mAH) , long life. Can Be used continuously for 10 hours on one charge.
- Unique circuit design, external charger as an AC power converter. The instrument is AC and DC, and can be charged and used at the same time.
WGG60-Y4 GLOSSOMETER APPLICATION:
Measurement of surface gloss of ink, paint, paint, paint and wood products.
BUILDING DECORATION MATERIALS: MARBLE, granite, glazed polished brick, pottery
Measurement of equivalent surface gloss of ceramic tile.
Surface gloss measurement of plastic and paper.
Surface gloss measurement of other non-metallic materials.
Technical parameters of WGG60-Y4 glossometer
Technical specifications, units, parameter values
Reading Range, gloss 0-199.0
Error in indication, unit of Gloss ± 1.2
Measuring Facula MM236X 18
Angle of projection, deg. 60
Volume (length x width x height) mm3130x 40x 80
The mainframe weighs 430 grams
Top of the form
Welcome to Dongguan Embellish Electronics Co. , Ltd. :
A brief description of the Florida gloss meter is as follows:
1. Measuring Range: 0-199.9 units of gloss
2. DIVIDING VALUE: 0.1 GU0.5 Gu1gu
3. Stability: ± 1 unit of Gloss
4. Zero error: 0 lustrous units 30 min
5. Indication error: ± 0.1 unit of Gloss
6. Power: Built in battery or direct use of No. 5 battery. At present, high-quality gloss meter manufacturers designed machine battery life up to more than 30 hours
7. Operating Voltage: 5V or dry cell conversion
8. Ambient temperature:-0 °C ~ + 40 °C
9. Storage ambient temperature-25 ° ~ + 40 °
APPLICATION OF GLOSSOMETER:
1. Measurement of surface gloss of ink, paint, paint, paint, wood products, etc. .
2, building decoration materials: MARBLE, granite, glazed polished tiles, ceramic tiles and other surface gloss measurement.
3. Measurement of surface gloss of plastic, paper, etc. .
Measurement of surface gloss of other non-metallic materials.
CHARACTERISTICS OF GLOSSOMETER:
1. No preheating is needed, and the measurement is stable for 3 ~ 5 seconds after starting up. Small and portable, switch with reading lock function, under-voltage display and charging indicator function.
2. The high-gloss Standard Plate is made of quartz single crystal material with an annual variation rate less than 0.3% , which has the stability of international advanced level.
3. Special Bulbs last for up to 10,000 hours.
4. Automatic Zero Correction, zero adjustment is not necessary.
5. Use nickel-hydrogen rechargeable battery, its capacity is large (100mAH) , long life. Can Be used continuously for 10 hours on one charge.
6. Unique circuit design with external charger as AC power converter. The instrument is AC and DC, and can be charged and used at the same time.
HOW TO USE A GLOSSOMETER:
1. Put The glossometer switch in the “OFF” position, take out the battery cover, install four ni-cd batteries according to the polarity, then press the battery cover.
2. pull the glossometer interface dial to the black standard edition values, dial a total of three, the first represents ten, the second represents a bit, the third represents a decimal point after a bit. If the standard block value is 93.3, then pull the dial to “933” .
Turn ON the power switch, in the “ON” position, at this time the LCD decimal point” . ” .
4. Place the glossometer measuring window on the pit (or Black Velvet) of the standard box, press the “zero” button and release it. After a few seconds, the glossometer will display “0” to indicate that the “zero” function is complete, please press the “zero” button again.
5. Place the measuring window of the glossometer on the Black Standard Board, press the “calibration” button once and then release it. The glossometer will display the dial value, indicating that the calibration function is completed.
6. Place the glossometer on a white ceramic plate and press the “measure” button. The glossometer will display other values. Note this value for future calibration instruments.
7. You can officially start measuring. Please press the “measure” button when measuring.
8. If the battery is low, the glossometer will display the “one” symbol, which indicates that the battery needs to be replaced.
9, to save electricity, if within 10 seconds, no key down, the display out. But in order to indicate that the instrument is in the boot state, the decimal point still shows that when a key is pressed, the glossometer will perform the key function.
Points to note when using a glossometer:
1. glossometers don’t need to be preheated.
2, in order to save electricity, if in the measurement process, if the longer time is not used, please turn off the power switch.
3, each time the power switch, must carry on “the zero calibration” and “the calibration mark” the step.
4. If not used for a long time, please take out the battery, because if the battery is kept in the electric appliance for a long time, it will leak because of its internal discharge.
5. The glossometer can use a nickel cadmium battery.
6. When the standard black edition is not in use, place it in a dryer.
7, the use of direct light should be avoided, or affect the accuracy of measurement.
Note: Gloss meter use No. 5 alkaline battery section, open the battery cover, the battery according to the positive and negative signs (see battery box) installed, please do not install the reverse. Change the battery in time when the under-power symbol “←” is displayed.
Dongguan Runzhi Electronics Co., Ltd.
Contact Person: Mr. Zheng
Department: Business Department
Post: Sales Manager
Telephone: +86 13712847488
Landline: +86 0769-81660769
Company Address: Room 101, 33 Zhenma Road, Dalingshan Town, Dongguan City, Guangdong Province,China
Factory Address: Room 101, No. 33 Zhenma Road, Dalingshan Town, Dongguan City, Guangdong Province,China
Previous: Muffle furnace
Next: Magnifying glass